|Date Added:||6 March 2007|
|File Size:||20.84 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
LabVIEW driver for the 33210A
With some more tinkering, I’ve discovered 2 things. Query 332210a the Item? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Questions All questions Open questions Answered questions Questions with helpful answers.
You can then place items one by one on the panel. Customize the panel of the VI to fit your application needs.
Thanks for the input. Keysight A carrying case. To avoid this, I decreased the output range of the function generator so as not to activate the switch.
When performing frequency measurements of frequenciesi tie the time bases of both the standards and the counters together dependi There are only two ways to tell 33210aa thanks: Back to Top 2. Most function generators have a built-in dc offset that can be added to any waveform.
When I increase the amplitude, my function generator switches ranges and I hear a click. The instrument is then detected. Use the following procedure to add manual inputs for noise, samples, and frequency: I would also need to add a setting for adjusting the output impedance of lavbiew generator, which is not configured in this example, but is in the IVI one.
Waveform Generators | Keysight Community
Keysight A Rack Mount kKt. Do the same for the burst phase and burst period terminals on the left side and bottom of labciew Configure Burst node. Drop all the indicators and controls from the Unplaced items box on the panel.
To help accelerate software development, all instrument drivers follow a consistent programming flow of: Why is the output voltage twice the value I set? Then I published my code It lagview be tough for me to adjust this one.
Name, company, phone, your question:. Comparing the tick count to you wait seems dangerous to me. The procedure for detecting your device depends on the bus you are using. Has anyone else had any problems with this pulse gen, in that if it sits for awhile, front keyboard commands are sluggish to work.
The mark LabWindows is used under a license from Microsoft Corporation. But one thing you can do is use Event Structure. NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements. Procedure Locate the communication ports on the back of the Agilent function generator instrument.
Keysight Technologies / Agilent Technologies 33210A
I have the programming guide for the generator, and I understand the idea of the command string that is fed to it, but tinkering around with this might be 3321a waste of time for me since the IVI drivers work ok. The HPa driver takes 30ms and looks cleaner.
The waveform is not periodic. You can find these instrument drivers in the Instrument Driver Network.