providing a window into the process. Rev: n/a. P/N: n/a. ECO: n/a. Dynisco 1/8 DIN Indicator. Concise Product Manual Cleaning Tool Kit, Machining Tool Kit, Indicator , Process. Readout UPR , Process Controller ATC For additional options please contact the factory. For additional options please consult factory. Accessories. Indicator , Process Readout UPR, Process Controller ATC,. Cleaning Tool Kit, Machining.
|Published (Last):||22 October 2011|
|PDF File Size:||19.19 Mb|
|ePub File Size:||3.51 Mb|
|Price:||Free* [*Free Regsitration Required]|
If anyone attempts to modify the parameters when the indicator is locked, the display will show inh. If your transducer was purchased in the US, it is preset at 10, psi, and no adjustment is necessary.
The Model can also be supplied with bidirectional and half duplex RS serial communications. The instrument parameters can be 190 by the master. Slide mounting clips fully forward with the y side toward the front until base is secure in panel.
At this point, be dtnisco that no pressure is applied and that the transducer is at operating temperature. Input Fail Safe followed by: If calibration is not needed, press FUNC to go to the next parameter.
– Pressure Indicator Manual in English_百度文库
HAr would indicate High alarm, automatic reset, reverse action. Then a process begins of applying successive the bytes of the message to the current contents of the register. You can display high and low peak values, lock and unlock the keyboard, reset alarms, and perform transducer input calibration and alarm threshold settings.
Easy-to-remember pushbutton sequences simplify transducer calibration routines.
Grizzly Bear 3D model. Default values are shown on page The two values must be equal. This error may appear in operating mode when storing new value in EAROM for example, alarm threshold or transducer calibration. Home Bedroom with Posh Floor 3D model. Reply Time The slave will start to send a reply from 2 ms to ms after dynixco end of the request detected by counting the received bytes. Resolution and decimal point position are as selected for the readout value.
A procedure for generating a CRC is: The system is now ready for use. Sanct Grael Van Eyck 3D model.
Alarm 1 hysteresis Alarm 2 operative modeRange: Permanently Connected Equipment A disconnectable device must be fitted with one of the following items: The increment is 1 for dynisci up to10 from tofrom to Press and hold FUNC until display returns to pressure reading.
The codes for the remaining digits in the alarm-operating mode are: To clear, initiate the default data loading procedure and repeat the configuration and calibration procedure.
Technical Data 9 5. A programmable voltage to current retransmission output is available as an option. The display will alternately show 1. Press the FUNC button to reach desired parameter then make output adjustments by pressing or until the signal output measured by the multimeter reaches the proper value.
Therefore, from the instrument front the parameters can be only displayed but not modified. If necessary, you may now select the analog ouput calibration procedure by pressing Analog Output Calibration, below. Higher temperature can result in damage and malfunction. Damaged instruments should be returned to: Only the numerical value will be shown during modification.
Continue doing this until all bytes have been processed.
Dynisco 1390 Indicator Series
Roulette Set 3D model. Automatic detection of maximum and minimum measured value.
The computer should be programmed to serve as a master controlling which slave has access to the link. Print Settings Printer Brand: In the package you will find: The perform calibration press The display will show tA Followed by: The content of this manual must be read, understood and followed in its entirety.
If you experience difficulties we are more than happy to give you qualified Monitoring Peak High and Peak Low By pressing while the measured value is displayed, it is possible to monitor the peak high value.
The display only shows the number of counts for the digital to analog converter. This error message disappears automatically after 3 seconds, and the instrument will reset. Remove the unit from the instrument case.